Does NI-DAQ support standalone deployment?
8 views (last 30 days)
So, here's the issue I'm having:
I've written and tested a GUI that reads from an NI USB-6009 adapter. This program works on my Windows 7 PC, with MATLAB both 2015A and 2014B, and I've installed the NI-DAQmx 14.0 driver.
The problem is, it's not working on the laptop that I ultimately have to deploy the application to. I compiled the GUI on my PC, and then installed it and the MCR on the laptop (running Windows 7), and the same NI-DAQmx 14.0 driver as well.
When I run the GUI as a standalone on my PC, it still works. But it doesn't even detect NI as a vendor when I try to run the standalone on the laptop, even though NI-MAX and Windows both detect the USB-6009 device. So, my question is, is there any incompatibility with the MCR and DAQ toolboxes that would cause this?
muhammet balcilar on 13 Nov 2015
Edited: muhammet balcilar on 15 Nov 2015
i had the same problem and i didnot solve with Matla2015, but i solved with using Matlab2014a. But Matlab 2014a had an other problem. First of all you have to comment out line 61 to line 65 in MATLAB\R2014a\toolbox\daq\daq\+daq\+ni\AnalogInputVoltageChannel.m file. it must be
% Set the Coupling
%[status] = daq.ni.NIDAQmx.DAQmxSetAICoupling(...
and then you have to compile all your code again with Matlab2014a. It will work.
Sagar Patil on 13 Oct 2017
I am trying to get NI DAQ support in my standalone application as well but not able to do so, not even with standalone deployed in my PC with MATLAB installed. Do we have to add any additional files during application compiler?
Sam Walder on 18 Feb 2019
I had the same issue in my application - the DAQ would connect and work in the full Matlab environment, but not in the compiled application. I resolved the issue by making sure to update everything in NI MAX on both the development and deployment machines. I am not sure why this was an issue - but it fixed it for me.
It may be useful to note that there are some similar questions on this topic on the forum: